Suzhou Kanglaituo Industrial Equipment Co., Ltd.
Address: No. 177 Pu Pu Road, Zhangpu Town, Kunshan City, Jiangsu Province
Environmental test equipment inside the most common temperature test equipment is the high and low temperature test chamber, a good equipment is indispensable is the daily maintenance, then the maintenance of high and low temperature chamber how much you know? Take you to understand the maintenance of high and low temperature test chamber Price:
1) the best use of high and low temperature test chamber temperature of 8 ℃ -23 ℃, do not have this condition of the laboratory, need to be equipped with appropriate air conditioning or cooling tower;
2) open the door or take the test object from the box, the goods must not be in contact with the door seal;
3) Compressor with air-cooled cooling, should regularly check the condensing fan, and the decontamination of the dust removal;
4) due to the cleanliness of different samples, in the forced air circulation under the action of the evaporator will gather a lot of dust and other small particles, so we must regularly clean the evaporator;
5) humidifier scaling easily lead to humidifier dry, may burn the humidifier, it must be regularly on the water and humidifier for cleaning;
6) At the end of each experiment, it is advisable to clean the test chamber with clean water to keep the equipment clean. The box should be cleaned after drying the box to keep the box dry;
7) adhere to the end of each test after the temperature set in the vicinity of the ambient temperature, work for about 30 minutes and then cut off the power, and the studio wall clean;
Summary of the daily maintenance of high and low temperature test chamber, the specific matter or need to refer to the instructions to use this product.